TE-4744, Tinytag Plus LAN, Ethernet Data Logger with four Voltage Inputs, 0-25VDC
TE-4744, Tinytag Plus LAN, Ethernet Data Logger with four Voltage Inputs, 0-25VDCThis data logger monitors voltage from 0 to 25V DC across four inputs and can be connected to a variety of sensors enabling the logging of a wide range of processes.
It plugs directly into a network point and sends data via the LAN for immediate viewing in the Tinytag Explorer Connect software.
The logger records data at user-defined logging intervals and can be programmed to give alarm alerts, both audible and visual, in the event of a problem.
The logger also has an alarm output that changes state when alarms are active.
The logger is supplied with: a PoE injector, an RJ45 shroud, four CAB-3239 voltage input cables and a detailed manual.
Alternatively you can buy four ACS-9703 5-pin connector plugs to connect your own probes/sensors.
LAN loggers can be mixed and matched with Tinytag Plus Radio and Ultra Radio data loggers within an application to meet particular site layout and infrastructure requirements.
The logger comes without 2 LAN cables for connection to the PoE injector. These need to be ordered separately.
Voltage input: 4
Measuring range: 0-25VDC
Logger resolution: better than 1mV
Max. input voltage: 26V
Input impedance: 100MOhm
Logger accuracy: 0.2% of reading ±0.02V
Logging intervals: 2 minutes to 10 days
Default logging interval: 10 minutes
Logging/off-line capacity: 361 days typical at a 10 minute logging interval
Alarms: 2 fully programmable, latching alarms
Alarm delay: 1 second to 10 days
Alarm indicators: visual (LED) & audible (sounder)
Alarm output: 3-pin, powered output that changes state when alarms are active
IP rating: IP67 (waterproof with waterproof shroud on the Ethernet connection fitted)
Operational range: -20°C to +55°C
Case dimensions (H x W x D): 161 x 113 x 62mm³
When used as a standalone device, a copy of the following software is required:
SWCD-0040 Tinytag Explorer
When used as a part of a Tinytag Connect system, a copy of the following software is required:
SWCD-0080 Tinytag Explorer Connect
CAB-0041 Alarm output cable for connection to an external alarm device