TE-4021, Tinytag Plus LAN, Ethernet Temperature Logger for one Thermistor Probe
TE-4021, Tinytag Plus LAN, Ethernet Temperature Logger for one Thermistor ProbeThis data logger monitors temperatures from -40°C to +125°C using one of the optional Tinytag thermistor probes. It plugs directly into a network point and sends data via the LAN for immediate viewing in the Tinytag Explorer Connect software.
The logger records data at user-defined logging intervals and can be programmed to give alarm alerts, both audible and visual, in the event of a problem.
The logger also has an alarm output that changes state when alarms are active.
The logger is supplied with: a PoE injector, an RJ45 shroud and a detailed manual.
LAN loggers can be mixed and matched with Tinytag Plus Radio and Ultra Radio data loggers within an application to meet particular site layout and infrastructure requirements.
The logger comes without a thermistor probe and without 2 LAN cables for connection to the PoE injector. These need to be ordered separately.
Reading range: -40°C to +125°C
Sensor type: thermistor (optional external probe)
Logger resolution: 0.02°C or better
Temperature stability: ±0.01°C/°C change from +25°C
Logger accuracy: 0.35°C (0 to +70°C) with a Tinytag thermistor probe
Logging intervals: 2 minutes to 10 days
Default logging interval: 10 minutes
Logging/off-line capacity: 361 days typical at a 10 minute logging interval
Alarms: 2 fully programmable, latching alarms
Alarm delay: 1 second to 10 days
Alarm indicators: visual (LED) & audible (sounder)
Alarm output: 3-pin, powered output that changes state when alarms are active
IP rating: IP67 (waterproof with waterproof shroud on the Ethernet connection fitted)
Operational range: -20°C to +55°C
Case dimensions (H x W x D): 161 x 113 x 62mm³
All thermistor probes which can be found in the Tinytag Radio Loggers categories.
When used as a standalone device, a copy of the following software is required:
SWCD-0040 Tinytag Explorer
When used as a part of a Tinytag Connect system, a copy of the following software is required:
SWCD-0080 Tinytag Explorer Connect
CAB-0041 Alarm output cable for connection to an external alarm device